the Creative Commons Attribution 4.0 License.
the Creative Commons Attribution 4.0 License.
Automated Static Magnetic Cleanliness Screening for the TRACERS SmallSatellite Mission
Cole J. Dorman
Chris Piker
Abstract. The Tandem Reconnection and Cusp Electrodynamics Reconnaissance Satellites (TRACERS) Small Explorers mission requires highfidelity magnetic field measurements for its magnetic reconnection science objectives and for its technology demonstration payload MAGnetometers for Innovation and Capability (MAGIC). TRACERS needs to minimize the local magnetic noise through a magnetic cleanliness program such that the stray fields from the spacecraft and its instruments do not distort the local geophysical magnetic field of interest. Here we present an automated magnetic screening apparatus and procedure to enable technicians to routinely and efficiently measure the magnetic dipole moments of potential flight parts to determine whether they are suitable for spaceflight. This procedure is simple, replicable, and accurate down to a dipole moment of 1.59 × 10^{3} N m T1. It will be used to screen parts for the MAGIC instrument and other subsystems of the TRACERS satellite mission to help ensure magnetically clean measurements onorbit.
 Preprint
(1329 KB)  Metadata XML
 BibTeX
 EndNote
Cole J. Dorman et al.
Status: final response (author comments only)

RC1: 'Comment on egusphere2022480', Anonymous Referee #1, 04 Aug 2022

CC1: 'Reply on RC1', Cole J Dorman, 19 Feb 2023
Referee
Journal of European Geosciences Union
DOI: https://doi.org/10.5194/egusphere2022480
Title: Automated Static Magnetic Cleanliness Screening for the TRACERS SmallSatellite Mission
Authors: Cole J. Dorman^{1, 2}, Chris Piker^{1}, and David M. Miles^{1}^{1}Department Physics and Astronomy, University of Iowa, Iowa City, 52242, USA
^{2}Department of Climate and Space Sciences and Engineering, University of Michigan, Ann Arbor, 48109, USA
Dear Referee,
We would like to thank you and for your careful consideration and time in handling our manuscript. We truly believe that the revised manuscript has been significantly improved by your suggestions.
The appendix below details our response to each of the comments. We hope that this revised and resubmitted manuscript addresses these comments appropriately. Please let us know if you have any questions regarding our resubmission. Thank you again for handling our manuscript.
Best Regards,

Cole J. Dorman
PhD PreCandidate
Department of Climate and Space Sciences and Engineering
University of Michigan, Ann Arbor
Email: cjdorman@umich.edu
Phone: (563) 2093148Appendix
Key:
Italics  Original Reviewer Comment
Bold – Author Response
(Text with Parenthesis) – Changed/Added Text
[[Text with Brackets]] – Large Changes to Manuscript, Not Detailed Here
No modifications – Original Text, Included for ContextLine 54, Figure 2, and line 97
It is better to specify the maximum size of an object to be tested. This point is related to an assumed farfield measurement.
RE: Thank you for the comment. We added further clarification in lines 103105 on the maximum sized objects to be tested:
“If the calculated dipole moment is less than its allocation then the measured object would be considered suitable to go on the spacecraft. (The farfield assumption of B relies heavily on the distance of the measuring sensors from the screening object. If the sensor is at least 5 times farther away from the centered screening objects characteristic radius, the farfield assumption holds (Bansal, 1999))”Figures 2 and 3
The subplots in Fig. 2 (times series measured by magnetometers at 11 cm and at 17 cm) are identical to the subplots in Fig. 3. This means that the authors can rearrange these figures to one figure.
RE: Thank you for your comment. While the subplots in Fig. 2 do reappear in Fig. 3, it is simply for continuity reasons for the reader. We believe Fig. 2’s schematic of collecting sinusoidal magnetic data using a spin modulated tray and Fig. 3’s demonstration of using discrete Fourier transform is distinct enough in meaning to be stay separated.
In the upperleft subplot of Fig. 3, the peaktopeak amplitude of Bz seems to be about or larger than 4000 nT, but the corresponding periodic amplitude in the upper right subplot is 3223.3 nT. Is this caused by a flattop window applied to time series? If it is the case, it is better to mention it. By the way, which is better, use of a flattop window or not?
RE: Thank you for the comment. An appeared change in periodic amplitude would be caused by the flattop window. We added clarification:
“(Due to the flattop window taking averages of magnetic intensity peaktopeak, it finds a periodic intensity of the sinusoidal peak divided by √2.)”
The flattop window lowers the frequency resolution of the Fourier transform in order to reduce amplitude noise. We believe we sufficiently explained the strength of using a flattop window over no window at all in lines 8185:
“Without a flattop window, slight changes in rotational frequency could disperse our target spinmodulated signal across multiple frequency bins in the DFT and degrade our estimate of the magnetic field component. A flattop window (D’Antona and Ferrero, 2005) is used to improve the accuracy of amplitude measurement at the expense of reduced frequency resolution (which is irrelevant in this application).”Equation (1)
The subscript m should be specified. Later, m is used as the magnetic dipole moment.
RE: Thank you for your comment. We decided to just remove the m subscript. It was not necessary and saves any confusion.Equations (2) and (3)
The subscript k should be specified.
RE: Thank you for your comment. We changed this subscript to be now a subscript “l”, as a k subscript appears unrelated later in the manuscript. The subscript l denotes the summation of variable x from l=0 to infinity and is common enough of mathematical notation that the reader will not need explanation. We also chose to remove the k subscript from ω.Equations (4) and (5)
Vectors r, ϑ, i, j, and k should be specified.
RE: We added additional information:
“Where B is the magnetic field, 𝜇0 is the vacuum permeability constant, m is the dipole moment, r is distance, θ is the dipole’s angle from the z axis, and 𝜑 is the dipole’s angle from the x axis. (Vectors r and ϑ denoted radial and azimuthal units in spherical coordinates, and vectors i, j, and k denote longitudinal, lateral and normal units in cartesian coordinates.)”
“ , ”between two equations for θ = ··· and φ = ··· is significant to separate these equations, so that add “comma” in Equation (5).
RE: We added a comma in between the two equations in Equation (5)
It is better to use different sign for the inner product of vectors, ·, and scalar multiplication.
RE: Thank you for your comment. We changed Equation (2) to use a “*” sign for scalar multiplication. Now every time a “·” sign is used, it is only for an inner product.Lines 106–110
“It is important that the object on the plate is centered.” I agree with it. However, a magnetic dipole in the object is not necessarily present at the center of the object. In the same sense, how about the height of magnetometers against the object? In other words, an offset dipole moment should be taken into account. This suggests that the present method may have any defect. To overcome this point, the authors can determine spherical harmonic coefficients up to degree 2…
RE: Thank you for the insightful comment and derivation. We have addressed concerns of a multipole expansion arising from a dipole moment not being geometrically centered below:
[[ We added an Appendix A after the conclusion to derive the multi expansion terms added to the magnetic scalar potential if the dipole moment is not geometrically centered in the object. The appendix serves to demonstrate if we express the magnetic field as a sum of multipole expansion terms and fit it to data, we can determine the offset and the dipole moment.
The length scale of the dipole, however, is much smaller than the measuring distance, so any higher order multipole terms fall off quickly enough to the point where they’re negligible. The negligibility of higher order terms preserves the integrity of the current screening method and is validated in Section 4, Validations. ]]
We now address a dipole offset in lines 113116:
“(If there is a dipole offset from the geometric center of the object, it will add multi expansion terms to the magnetic scalar potential as elaborated in Appendix A. However, the distances we are measuring the magnetic fields are much larger than that of the length of the dipole and hence multipole terms fall off quickly enough to be negligible for the scope of this project.)”Lines 125–126
“It ··· are centered ···” would be “It ··· is centered···.”
“··· a 40×40 cm cubic ···” would be “··· a 40×40×40 cm cubic···.”
RE: Thank you for the comment. We made the appropriate changes in the manuscript.
“It, along with 2+ Twinleaf VMR magnetometers for data collection, (is) centered in a 40 x 40 (x 40) cm cubic mumetal magnetic shield”Lines 144–147
The authors describe that there is a vertical alignment error as one of errors. This can be reduced if the position of a magnetic dipole moment is simultaneously determined as mentioned above.
RE: Thank you for the comment. We believe your response to comments on Lines 106110 is sufficient enough where this critique is no longer an issue.Figure 6
The red and black cables are likely to be used as power lines. Are they twisted? If it is not the case, such a configuration may cause additional magnetic field, so that they should be twisted. If it is the case, it is better to point out the configuration.
RE: The wires drawing power are twisted and we added an explanation in the Figure 6 caption to better reflect that:
“Figure 6: Independently characterized solenoid that was used to validate the automated magnetic screening apparatus and procedure. (The wires drawing current from the battery are twisted to minimize additional magnetic fields.)”Equation(6)
The subscript k should be clearly defined. If k stands for x, y or z, the lefthandside of Equation (6) should be mk, where m = (m^{2}_{x }+m^{2}_{y} +m^{2}_{z})^{1/2}.
RE: Thank you for the insight. The dipole, m, effectively only has a zcomponent. Since Line 97 now denotes “(vectors i, j, and k denote longitudinal, lateral and normal units in cartesian coordinates)” and we removed the k subscript from Equations (2) and (3), the reader is confidently informed the k subscript defines the z or normal coordinate unit. We changed Equation (6) to include a k subscript in the lefthand side of the equation, to address your comment and better reflect the system being measured.Citation: https://doi.org/10.5194/egusphere2022480CC1

CC1: 'Reply on RC1', Cole J Dorman, 19 Feb 2023

RC2: 'Comment on egusphere2022480', Anonymous Referee #2, 05 Sep 2023
Achieving a 100 nT cleanliness at 1 m would be achievable using simpler screening equipment. A magnetometer in a magnetic screen would be sufficient. An even easier way is to use an astatic magnetometer that measures the gradient of the dipole moment directly.
The described method can however be used for more strict requirements and resembles the Multi Dipole Model method used eg. in MFSA of IABG, Germany to comply with ECSSehb2007a
Figure 3 On the data processing
The peaktopeak amplitude of the blue AC trace seems to be 2500 nT while the magnitude in the frequency domain is calculated to 976.3 nT. It seems that the flattop window reduces the peak determination by some 20%. This should be discussed.
Almost 7 full periods are used in the DFT. Using a few periods may significantly influence the magnitude determination in the frequency domain. Please discuss the tradeoff between test time, rotation speed sample rate and signal processing (averaging spectrum).
Line 88 “to produce an averaged spectrum”:
This is not described in the procedure section below. How many spectrums are averaged? What noise reduction is achieved by this?
Equation 4
A figure describing the reference frame the dipole moment and the dipole field would help.
Line 98 “the magnetic field vector completely aligns with the dipole moment vector"
Is that true? I guess the direction of the dipole moment can be determined by the measurements performed but the field vector is not always aligned with the dipole moment vector.
Line 110 “The plate is centered in a magnetic shield to reduce, but not completely remove, the background magnetic fields”
The shielding also reduces induced fields from the test object.
Line 106 “It is important that the object on the plate is centered”
It is important that the apparent magnetic moment of the test object is centered. This is together with the apparent height of the dipole an error source and should be evaluated.
Line 118 “Similarly, when the dipole axis and the spin axes are near parallel”
I guess "near parallel" should be "near perpendicular" but then I do not see the point.
Line 147 “These contribute to the error on each calculation of the worstcase fields”
What is the estimated combined error? The error bars in Figure 5 seem not to include all error contributions. I would expect larger error bars at smaller distances.
Citation: https://doi.org/10.5194/egusphere2022480RC2
Cole J. Dorman et al.
Cole J. Dorman et al.
Viewed
HTML  XML  Total  BibTeX  EndNote  

343  160  22  525  15  14 
 HTML: 343
 PDF: 160
 XML: 22
 Total: 525
 BibTeX: 15
 EndNote: 14
Viewed (geographical distribution)
Country  #  Views  % 

Total:  0 
HTML:  0 
PDF:  0 
XML:  0 
 1